__TIPS AND TRICKS IN FIELD INSTRUMENTATION__

**For **__RTD Pt 100__ measurement,measure the resistance across the white and common terminal, then the temperature can be calculated simply by

Temp=(resistance measured across terminal minus 100)/0.385 or (resistance measured across terminal minus 100)multiplied by 2.6

for example

if the resistance across white and red terminal is 126 ohm, then

the temperature measured is 26/0.385=67.53 degree centigrade or 26 multiplied by 2.6

26*2.6=67.53

This follow the same eqn, R=Rₒ(1+αT) remember that this is only applicable for PT 100 not other types

__For calibration of -100 mmH2O to -10 mmH20 range capillary type using pressure pump not vacuum pump __

The values for 25%,50%,75%,100% are as follows:

0%————- -100 mm h2o

25%———– -77.5 mm h2o

50%——– -55 mm h2o

75%——— -32.5 mm h2o

100%—— -10 mm h2o

First find the Span=URV-LRV=-10+100=90

Then divide this by 4 as we are calibrating for 4 values namely 25%,50%,75%,100%

i.e. 90/4=22.5

then the 4 points can be calculated as follows

0%(4ma)————- 0 mm h2o i. e LP and HP open to atmosphere

25%(8ma)———– -0+22.5=22.5 mm h2o (apply 22.5mmh2o to HP side not LP here LP is open to atmosphere.)

50%(12ma)——– 22.5+22.5=44 mm h2o (apply 44mmh2o to HP side not LP)

75%(16ma)——— 44+22.5=67.5 mm h2o (apply 67.5 mmh2o to HP side not LP)

100%(20 ma)—— 67.5+22.5=90 mm h2o (apply 90 mmh2o to HP side not LP)

__Calculation of flow m3/hr from differential pressure values mm h20 if both ranges are known__:

We know that the flow equation is related as follows

Q=k√∆p

Here Q is the rate of flow: k is the Bernoulli’s constant; and ∆p is the differential pressure

Consider for instance the D.P. transmitter is of range 0 to 120 mm H2O and the DCS range of

0 to 1500 m3/hr

Then the next step is to find the Bernoulli’s constant

i.e. Q=k√∆p

1500=k√120 (here we consider span URV values to find Bernoulli’s constant)

k=1500/√120

k=136.936

Once we get Bernoulli’s constant we can calculate any flow rate if we know the D.P.

For e.g.

If differential pressure is 90 mmH2O

Q=k√∆p becomes

Q=136.936√90

= 136.936*9.486 = 1298.9 m3/hr

Thus we can calculate any flow rate if we know the transmitter and DCS range.

4.For K type thermocouple (Chromel alumel) if the mV measured across yellow(positive) and red (negative) is x,then the temperature can be calculated as follows

Temperature=x/0.0397(millivolt measured divided by 0.0397) or x *25.2 (millivolt multiplied by 25.2)

for example if we measure the mV value across yellow and red terminal using a multimeter and found to be 0.397 then temperature can be calulated by

temperature=(0.397/0.0397)=10 degree centigrade or (0.397*25.2)=10 degree centigrade

6.calibration checking of capillary type LT if you dont have any instruments for checking

Suppose that a capillary type LT is mounted on a tank having range of -1200mmh2o as LRV and URV -60 mmh2o and you need to check whether the transmitter is Ok

We know that the transmitter is mounted with HP side to high pressure side and LP tapping to low pressure side,Firstly isolate the process line,vent and drain the process inorder to release any trapped pressure inside the flange.the transmitter will show 0% reading ie(-1200mmh2o),

Now measure the tap to tap length and mark the corresponding 25%,50%,75% and 100% level .Remove the LP flange of the transmitter(with HP flange of capillary intact) and keep it near(parallel) to HP tapping, the transmitter will show 100%.Now lift the LP capillary flange to 25% above from the HP tapping (where we marked before as 25%),now the transmitter will show 75% (not 25%)

Next keep the transmitter LP capillary flange at 50% marking the transmitter will show 50% reading.Similarly when we place at 75% the transmitter will be showing 25%

Finally if we place at 100% marking the transmitter should show 0% that is -60mmh2o